You are here: Foswiki>DaqSlowControl Web>TDCReadoutBoard>TDCReadoutBoardV3>BookkeepingTrb3 (2015-02-02, MichaelTraxler)Edit Attach

Serial Number | current Location | Flash with 0R | Errors, Bugs, Special Features |
---|---|---|---|

1 | Frankfurt | X | patch wire fixed SFP1 fixed |

2 | Frankfurt | X | |

3 | Mainz | ||

4 | Cracow | ||

5 | Mainz | ||

6 | GSI | FPGA 5 only | with external linear power supply |

7 | Frankfurt | - | no communication with FPGA 2 |

8 | Mainz | no access to Flash 1 - fixed | |

9 | Mainz | DCDC4 (1,8 V) doesn't work - not fixed no access to Flash 4 - fixed | |

10 | ? | not existing board? | |

11 | Erlangen/WASA | x | |

12 | Mnchen | x | all 4 Samtec-connectors have no contact on 3.3V and 6V |

13 | Gieen, Benno Kroek | Seems that the 3.3V are missing on the AddOn -Connectors | |

14 | KVI | x | only lent, no errors known |

15 | GSI | x | Edge AddOn4: power blades are not connected |

16 | Erlangen/WASA | x | |

17 | Mainz A2 | x | |

18 | Mainz A2 | x | |

19 | Mainz A2 | x | |

20 | Erlangen/WASA | x | |

21 | GSI | x | |

22 | Erlangen/WASA | x | |

23 | Erlangen/WASA | x | |

24 | Erlangen/WASA | x | |

25 | Erlangen/WASA | x | |

26 | Erlangen/WASA | x | |

27 | Erlangen/WASA | x | |

28 | GSI | x | |

29 | Wuppertal | x | |

30 | GSI | x | |

31 | Krakow/Marek | x | |

32 | Krakow/Marek | x | |

33 | Munich/Rafal | x | |

34 | GSI | x | |

35 | Jerusalem/PSI | x | Lent to Guy Ron |

36 | Mainz | x | |

37 | Mainz | x | |

38 | Mainz | x | |

39 | Mainz | x | |

40 | Mainz | x | |

41 | KVI | x | |

42 | GSI | x | Lent to Guy Ron |

43 | GSI | x | Evie Downie, Washington, only lent |

44 | Heidelberg | x | Christian Simon, Uni HD |

45 | GSI | x | NA61, Jovan Puzovic, Belgrade |

46 | GSI | x | |

47 | GSI @Wu | x | RESET_OR doesn't work, not possible to reboot central FPGA |

48 | GSI | x | |

49 | GSI | x | |

50 | GSI | x | |

51 | GSI | x | |

52 | GSI | x | |

53 | GSI | x | |

54 | GSI | x | |

55 | GSI | x | |

56 | GSI | x | |

57 | GSI | x | |

58 | GSI | x | |

59 | GSI | x | |

60 | GSI | x | Mainz - lent for the July2013 test beams |

61 | GSI | x | Mainz - lent for the July2013 test beams |

62 | GSI | x | Mainz - lent for the July2013 test beams |

63 | GSI | x | Mainz - lent for the July2013 test beams |

64 | GSI | x | back to GSI, DCDC2 broken, short circuit, repaired |

65 | Mainz | x | A2, Andreas Neiser |

66 | Mainz | x | A2, Andreas Neiser |

67 | Mainz | x | A2, Andreas Neiser |

68 | Mainz | x | PANDA, Tobias Wagner |

69 | Frankfurt | x | |

70 | GSI | x | can't reach FPGA 1&2 with slowcontrol |

71 | Erlangen/WASA | x | |

72 | GSI | x | |

73 | GSI | x | Two temperature sensors stop working higher than 30 degrees. |

74 | Frankfurt | x | BCM one pin not soldered(SOLVED). 2 temperature sensors not working. |

75 | GSI | x | |

76 | Giessen | x | |

77 | Giessen | x | |

78 | GSI | x | |

79 | Giessen | x | |

80 | GSI | x | |

81 | GSI | x | Pion Tracker, cave |

82 | GSI | x | |

83 | GSI | x | |

84 | GSI | x | |

85 | GSI | x | |

86 | GSI | x | Short circuit by BCM, will be sent back to Visatronic for repair |

87 | GSI | x | Power supply error - Overload/overcurrent LED is on - possible short curcuit |

88 | Frankfurt | x | |

89 | GSI | x | |

90 | Frankfurt | x | |

91 | GSI | x | |

92 | GSI | x | |

93 | GSI | x | |

94 | GSI | x | No DHCP req. after programming |

95 | GSI | x | |

96 | GSI | x | |

97 | GSI | x | |

98 | GSI | x | DONE3 LED does not glow |

99 | GSI | x | |

100 | GSI | x | |

101 | GSI | x | |

102 | GSI | x | |

103 | GSI | x | |

104 | |||

105 | |||

106 | |||

107 | |||

108 | |||

109 | |||

110 | |||

111 | |||

112 | |||

113 | |||

114 | Heidelberg | Christian Simon, Uni HD | |

115 | Heidelberg | Christian Simon, Uni HD | |

116 | Heidelberg | Christian Simon, Uni HD | |

117 | |||

118 | |||

119 | Heidelberg | x | Christian Simon, Uni HD |

120 | Heidelberg | x | Christian Simon, Uni HD |

121 | Heidelberg | x | Christian Simon, Uni HD |

122 | GSI | x | |

123 | GSI | x | |

124 | GSI | x | |

125 | GSI | x | |

126 | GSI | x | Initial problems with DHCP offer during programming, but problem seems to be solved |

127 | GSI | x | |

128 | GSI | x | |

129 | GSI | x | |

130 | GSI | x | Christian Simon, Uni HD |

Edit | Attach | Print version | History: r53 < r52 < r51 < r50 | Backlinks | View wiki text | Edit wiki text | More topic actions

Topic revision: r53 - 2015-02-02, MichaelTraxler

Copyright © by the contributing authors. All material on this collaboration platform is the property of the contributing authors.

Ideas, requests, problems regarding Foswiki Send feedback | Imprint | Privacy Policy (in German)

Ideas, requests, problems regarding Foswiki Send feedback | Imprint | Privacy Policy (in German)