Here a list of all taken hld-files should be maintained.
All files are to be recorded in: hadaq@lxhadeb01:/data01/data
MDC I Sector 0
MDC I Sector 1
MDC I Sector 2
MDC I Sector 3
2011-03-31: Test of MDCI Sector 3. Used file: te1109010143001; threshold 40
MDC I Sector 4
MDC I Sector 5
2011-03-03 16:45: Motherboard 8 Exchanged (File after change:te1106216455301)
All TDC of MB8 are appering in calibration polt.
2011-03-03 17:45 : Motherboard 2 Exchanged (file: te1106217411001)
2011-03-03 18:15 : Motherboard 2 Changed Back (file: te1106218302301)
Motherboard 4 Exhcanged
2011-03-03 18:15 : Motherboard 9 Changed Back (file: te1106310434401)
2011-03-15 16:55 : Module 0, sector 5 tested (file: te1107416113801)
(L. Rehnisch, H. Schuldes, this is NOT our Wiki account)
- noise occupancy:
- noise occupancy (old):
Comparing the noise occupancy to an old plot, we conclude, that there are no hot spots (the enhancement in MB 3 was because Attilio was touching this board to increase the noise).
Calibration and Slopes are fine.
Following sectors are tested: 2,4,5 (computing from 0)
Following sectors are tested: 0, 2,4,5 (computing from 0)